It’s a saying we’ve all heard – our gas is clean and dry. For many years that may have been the case but all of a sudden the regulators have frozen off, or the sample probes are clogged, and the Gas Chromatograph is in alarm and not providing results due to no gas flow. You are puzzled. How could this be happening if the gas is clean and dry?
This may be a story you know all too well.